Characterization of a Field-effect Transistor (FET) using an EmStat Pico potentiostat module.
The EmStat Go is a battery powered handheld potentiostat with the logo of your company. It can be fully customized to enable many electrochemistry applications. Read more drop det...
The PSExampleApp is a simple open-source application that guides a user to do a measurement with a PalmSens device to measure concentrations in a liquid sample. The app can be re-configured without code for use with (bio)sensors measuring a specific analyte using a linear calibration curve.
Learn how to use a multi-channel potentiostat as a polypotentiostat, so you can use multiple working electrodes in the same cell sharing one reference and counter electrode.
This article will explain how to update the software of your EmStat Pico Development Board, Sensit BT or Sensit Smart using PSTrace. ...
Get more insight into electrochemistry by adding a spectrometer to your potentiostat. This application note describes how to perform spectroelectrochemistry with a PalmSens4 and an Avantes potentiostat.
The specifications of potentiostats contain many parameters. If you are in doubt about the meaning of a certain parameter, worry no more. Below you will find an explanation for the...
This tutorial describes how to change the settings of the LAIRD BT900 Bluetooth module on PalmSens devices for OEM.
This application note provides a comprehensive set of instructions for use of the screen-printed electrode (SPE) IS-HM in analyzing Pb, Cd, and Cu concentrations of a polluted river water sample.
A comprehensive set of instructions for use of the screen-printed electrode (SPE) IS-Au for determining the Hg concentrations of aqueous solutions.
This application note is used to verify the IS-C (graphite) sensors and their electrochemical stability.
The screen-printed carbon electrodes by ItalSens IS-C have been characterized by measuring naphthol, and can also be used for quantification of naphthol.
The multiplexer MUX8-R2 offers the possibility to record reliable Open Circuit Potential (OCP) measurements in the Alternating mode, which allows to monitor the OCP of multiple samples / working electrodes over time with a single potentiostat. This is a significant improvement compared to its predecessors MUX8 and MUX16. In this application note it is demonstrated how to setup and perform an OCP measurement with the MUX8-R2 in the Alternating mode.
In this application note, the influence of noise on a measurement using a differential pulse technique is demonstrated.
This application note describes how to create a calibration for quantitative measurements of vitamin C. Due to the detailed description and harmless solutions this an excellent electrochemical experiment with liquids for beginners in the field of electrochemistry.